|PH:||NSS ACSS6.5~7.2 CASS 3.0~3.2||Power:||1A C220V|
|Certificate:||ISO9001:2008||Design Criteria:||CNS, JIS, K5400, Z2371, ASTM|
|Spray Quantity:||1.0~2.0 Ml / 80cm2 / Hr||Warranty Period:||12 Months|
|Brine Temperature:||35°C±1°C||Test Chamber Temperature:||NSS ACSS 35±1°C CASS:50±1°C|
corrosion testing equipment,
salt spray test machine
Airflow Spray Tower Corrosion Testing PID Fuzzy Control Neutral Corrosion Test Chamber
HD-E808-60 Salt Spray Chamber.pdf
HD-E808-90 Salt Spray Chamber.pdf
HD-E808-120 Salt Spray Chamber.pdf
Airflow spray tower Corrosion Testing PID fuzzy control Neutral corrosion test Chamber is Mainly used for the determination of inorganic coating of metal or non-metal or organic coating corrosion resistance test, mechanical research, solar photovoltaic, light industrial electronics, instrumentation and other industries a variety of environmental adaptability and reliability of an important test equipment
Do Test Method:
Neutral salt spray test (NSS test) salt spray test (SS test)
Acetic acid salt spray test (ASS test)
Copper accelerated vinegar instinct test
As a neutral salt spray test (NSS test), reinforced with PVC rigid plastic board when the salt spray test (SS test).
Using as acetic acid salt spray test (ASS test), copper accelerated vinegar instinct test (CASS test) when the PP plate.
Compared PP board PVC board and more resistant to corrosion at high temperatures.
1, Box material: The machine uses high temperature and humidity inside and outside the box are highly corrosive Class A gray PVC (thickness 5 ~ 8mm), the use of advanced three-dimensional reinforcement ring with a production technology, welded, strong structure, solid never deformation, gorgeous and elegant, suitable for salt spray, copper acetate, and other test specification test.
2, reagent bottles added: Hide-one supplement reagent bottles, easy to clean, easy to operate.
3, the pressure saturation tank: use SUS304 # stainless steel, fine pot shaped structure design, high temperature, high humidity, high pressure, highly corrosive, and after several years of accumulated experience, the successful resolution of the general market pressure salt spray machine short barrel water technical problems, a unique internal design pressure drums,
|Inside box size(cm)||60x45x40||90x60x50||120*50*100||120x80x50|
|Outside box size(cm)||113*107*63||146*128*91||175*120*148||224*150*160|
|Testing room temperature||Salt test (NSS ACSS)35±1 degree / Corrosion Test (CASS)50±1 degree|
|Pressure barrel temperature||Salt test (NSS ACSS)47±1 degree / Corrosion Test (CASS)63±1 degree|
|Brine temperature||35±1 degree or 50±1 degree|
|Testing room Capacity||108L||270L||600L||800L|
|Brine tank capacity||15L||25L||30L||40L|
|Salt concentration||The concentration of sodium chloride 5% or the 5% concentration of sodium chloride add 0.26g per liter of copper chloride (CuCl2 2H2O)|
|Spray volume||1.0~2.0ml/80cm2/h (working at least 16 hours, and then take the average)|
|Testing room relative humidity||85% above|
|Spray form||Programmable spray (Including continuous and intermittent spray)|
CNS, 3627,3885,4159,7669,7670,8886,JIS,D020,H8302,H8610,K5400, Z2371,ISO,3768,3769,3770,ASTM,B117, B268
1. Spray device: spray tower using a nebulizer (spire height adjustable), guiding salt spray, fog particles small, natural subsidence; without nozzle salt crystals, tunable settlement
2. Spray gas treatment: two spray gas regulator regulator, the same period the oil to be filtered, humidified gas preheating
3. fogging treatment: spray brine stored as hidden and built a large storage capacity, brine with preheating function
Saline spray before brine with quartz filter element, to avoid blocking the nozzle of impurities Sergio terminal test
4. Pressure control: by regulating valve inlet pressure, spray pressure regulating valve, pressure gauge
Air pressure is 1Kg / cm2 adjusted in two stages
Period of adjustment for the roughly 2Kg / cm2, imported air filters, with drainage function
Sec for the fine adjustment 1Kg / cm2,1 / 4 pressure gauge, display precise and accurate
5. Pressure Value: front air pressure: 0.3-06mpa
After the spray pressure: 0.07-0.17mpa
6. Air compressor: Buy requirements: exhaust pressure: ≥8 kg volume flow: ≥0.12m3 / min
Integrally molded at high temperature welding, corrosion-resistant, easy to clean, no leakage phenomenon.
Tower spray system, and equipped with saline filtration system, no crystal nozzle, spray evenly distributed free to adjust the amount of the settlement.
It covers the use of transparent materials can be clearly seen inside the test items and spray situation.
Using a water seal between the lid and the box structure, without salt overflow.
Strength line control panel and other parts are fixed in position to facilitate inspection and maintenance, the use of door-opening side door, not only beautiful, but also easy to maintain.
1, regular cleaning and maintenance of salt spray chamber pressure tank, brine tank nozzle.
2. Do not pressure sodium chloride solution from the barrel manually add water poured out, the first transfer solution within the cylinder and then further cleaned with distilled water.
3, Do not test the exhaust pipe behind the blockage, in order to avoid too much pressure to make the cabinets housing deformation.
4, do not put inflammable and explosive materials inside.
5, salt spray chamber behind the pressure gauge is 2 kg, 1 kg in front, factory in Wuhan still test test
Contact Person: Mr. Liang