|Frame Capacity:||200kg||Load Cell Brand:||China Brand, 200kg (5,10,50,100kg Are Optional)|
|Load Accuracy:||1% F.S||Motor:||Adjustable-speed Motor, Ball Screw Rod|
|Resolution:||Adjustable-speed Motor With Reduction Box, High-precision Mechanical Ball Screw Rod||Control System:||Controller With LED|
|Force Reading:||KG/N/LB||Vertical Test Space:||900mm Including Fixture|
|Test Speed:||50~300mm/min (adjustable By Rotary Knob)||Software:||TM 2101|
Laboratory Computerized Tensile Test Apparatus,
200kg Capacity Tensile Test Apparatus,
300mm / Min Tensile Test Machine
Laboratory Computerized Automatic Single Column Universal Tensile Test Apparatus
1. Tested sample: plastic films, laminated materials, adhesive tapes, adhesive bandage (plaster), release paper, plastic films, leather, rubber & plastic, paper, fiber, etc.
2. Different test with different grips: tensile, peel, tear, heat seal, adhesive, bend and open force.
3. Deformation under defined load: Test deformation under defined load of tested sample
4. Load under defined deformation: Test load under defined deformation of tested sample
5. Software can issue word/excel report with results for max. force, elongation, tensile strength, peel strength, tear strength, compression strength, etc.
This machine is a kind of electric tensile tester which uses the motor to drive ball screw to move fixture. Place the fixture with sample between the upper and lower fixture, use a given speed to pull sample upward by upper fixture. Load cell above upper fixture will sense tensile strength, and convert the strength into voltage sign and output to the display screen. And the strength value will be displayed automatically.
|Load cell brand||China brand, 200kg (5,10,50,100kg are optional)|
|Load accuracy||1% F.S|
|Control system||Controller with LED|
|Motor||Adjustable-speed motor, ball screw rod|
|Vertical test space||900mm including fixture|
|Test speed||50~300mm/min (adjustable by rotary knob)|
|Data display||Max. break force|
|Safety features||Over-load protection
Upper and lower limit switches
Load sensor with automatic retreat
1. Motor system:AC motor +Driver +High precise ball screw (Taiwan)
2. Control system: Microcomputer
3. Displacement resolution: 0.001mm
4. It is with build-in printer, can print the testing date, time, maximum, average and minimum value.
HAIDA TEST EQUIPMENT TEST STANDARDS FOR SHARING
Adhesives Tape/Pressure Sensitive Tape
EN 1719: Loop Tack Test
ASTM D3654: Shear Adhesion Test
ASTM D3759: Tensile &Elongation Test
FTM: 180&90 Degree Peel Test, Loop Tack
EN 1939, ASTM D3330: 180&90 Degree Peel Test
ISO 11339, ASTM D1876, ASTM F2256: T-type Peel Test
ASTM D903: Peel & Stripping Test
ASTM D1002: Shear Strength (Adhesive for metal to metal)
ISO 4587: Tensile lap-shear strength (Rigid-to-rigid bonded assemblies)
ASTM D882: Tensile Test
ASTM F1306: Puncture Test
ASTM D1938: Trouser Tear Test
ASTM D2732: Thermal Shrinkage Test
ASTM D1709, ISO 7765-1: Free-Falling Dart Method
ASTM D1894: Static and Kinetic Coefficients of Friction
ASTM D1424, ASTM D1922, ISO 1974: Elmendorf Tearing Strength Test
Plastic& Plastic Pipe
ISO 180: Izod Impact Test
ISO 9966: Ring Stiffness Test
ISO 179: Charpy Impact Test
ISO 4892: Weathering Aging Test
ASTM D3163: Plastic Lap-Shear Test
ISO 4422, ISO 3127: Pipe Impact Test
ASTM D4565, ISO 11357, EN 728: DSC/OIT
ISO 75: Heat Deflection Temperature (HDT)
ASTM D1598: Hydrostatic Pressure and Burst Test
ISO 1133, ASTM D1238: Melt Flow Rate Test (MFR/MVR)
ASTM D790, ISO 178, ISO 14125, ASTM D6272: Bend Test
ISO 306, ASTM D1525: Vicat Softening Temperature (VST)
ASTM D882, ASTM D 638, ISO 527, ASTM D1708: Tensile Test
1. Attentions: the machine cannot be used following situations:
1.1 Vibration, rocking place.
1.2 Direct sunlight.
1.3 Hot, dusty, damp places.
1.4 To ensure safe, AC supply of the machine should be well grounded.
1.5 Do not use strong solvents (such as: benzene, nitro oil) washing machine.
3.6 Do not inject water and debris into the machine to prevent damage to electrical components and electrical shock.
3.7 Machine’s disassembly and debugging can only be measured by the State Department approved the units and the company, other people not allowed to overhaul.
Contact Person: Mr. Liang