|Usage:||Multifunctional Optical Measuring Machine||Using Area:||Machinery, Electronics, Instrumentation, Plastics|
|Warranty:||1 Year Free||Image:||Clear|
|Feature:||Easy To Operate||Price:||Reasonable|
|Camera:||Color 1/3``CCD Camera||Optical Magnification:||0.7~4.5X|
Coordinate Measurement Machine,
Optical Measurement Equipment
2D Optical Electronic Measurement Coordinate Measuring Machine Video
Programmable 2D Video Electronic Measurement Optical Measuring Machine is designed to measure the two coordinates in all application areas and is widely used in different kinds of precision finishing industries, such as: machinery, electronics, ironware, instruments, plastics and so on.
Features of 2D Video Electronic Measurement Optical Measuring Machine:
1 > Creative modern-shape-design;
2 > Fashionable auto-painting-craftwork ;
3 > Steady granite base;
4 > Stable imported V-shape guide;
5 > Precise metal and glass worktable;
6 > High-speed 3-axis CNC driven motor;
7 > Adjustable USB 40-division ring LED illumination;
8 > Advanced Pomeas 0.7~4.5X Zoom lens;
9 > Wonderful SONY 1/2`` Color CCD camera;
10> Powerful 2.5D measuring software QuickMeasuring 4.1 version.
Position, concentricity, straightness, etc.
The Main Technical Parameter:
|Model||TT-U801Video Electronic Measurement Optical Measure Machine|
|Glass table size||320×280mm|
|Net Weight||240 kg|
|Travel: Z : 150mm||Work distance: 95mm|
|Video system||Camera: Color 1/3``CCD camera|
|Contour and surface illumination: Ring LED and brightness can be setup|
|Software||Quick Measuring 2.0|
Structure and composition:
Image measuring instrument is a kind of high-resolution CCD color camera, continuous zoom lens, color displays, video cross line generator, precision optical ruler, multi-function data processor, 2D data measurement software and high-precision table structures precision optical image measuring instrument.
The basic measurement principles:
(1) Abbe principle: in principle for the measured length and the reference length during the measurement should be placed on the same straight line. If the measured length and the reference length placed side by side comparison of the measurement process because of a manufacturing error occurs between the angle, offset, direction of movement, and have a greater length of two errors. Size of the error in addition to the angle between the size and the length of two related, but also its distance between the size, the greater the distance, the greater the error.
(2) the reference harmonized principles: unity and processing of measuring reference to the use of benchmarks and reference. That step measurement process should reference measuring standard, the final amount should be designed to detect the reference measuring standard.
1, the instrument should be placed in a clean, dry room, avoid the optical surface of the part sweat loss, metal parts to rust, dust and debris falling into the movable rail, affecting the performance of the instrument.
2, the instrument after use, should be ready to wipe clean working surface, and then cast the best guard.
3, transmission equipment and rail movement, should be regularly lubricating oil.
4, glass and painted surfaces dirty, you can wipe clean with a mild detergent and water. Never use organic agents to clean the paint surface, otherwise it would tarnish the paint surface.
5, precision instrument parts, such as the impact of the system, table, optical device and Z-axis drive mechanism etc. required fine adjustment, all adjustment screws and sturdy screws are fixed and can not disassemble.
Programmable 2D Video Electronic Measurement Optical Measuring Machine view:
Contact Person: Mr. Liang