|Humiture Range:||30-98%RH||Rainfall Cycle:||1-240min|
|Certificate:||ISO9001: 2008||Warranty Period:||12 Months|
|Studio Size:||95Dx95Wx90Hcm||Box Material:||Stainless Steel SUS304#|
|Insulation Materials:||Superfine Glass Fabric||Control System:||Germany KLOCKNER-MOELLER Progammable Controler|
|Gas Processing:||Dischager For Fector And Harmful Gas||Use Environment:||5°C~+30°C, ≤85%R.H|
ozone test chamber,
xenon test chamber
Professional Xenon Test Chamber Laboratory Testing Equipment / Instrument
Xenon Lamp Aging Chamber is to promote the deterioration of vulcanite, and then to calculate the changerate oftensile and elongation. Generally consider being tested one day in the 70 degree is theoretically equal to 6 monthsexposed at the outside atmosphere.
Door handle: no reaction door handle, easier to operate
Moves the sample holder: Sample holder rely installation box at the bottom of the linear reciprocating drive system for moving up and down
Casters: bottom of the machine can be fixed with high quality PU wheels
Air circulating means
Built-in air-conditioned rooms, circulation air duct and the long axis of the centrifugal fan, the use of energy efficient refrigerators and conditioning systems, through efficient fan for effective exchange of the purpose of the temperature change. By improving the flow of air, improve air flow and the heater and air cooler heat exchange capacity, thereby significantly improving the temperature uniformity of the chamber.
|Model||HD-711L Xenon Lamp Aging Chamber|
|Inner Box Size (D×W×H)||950×950×900mm|
|Outside Dimension (D×W×H)||1300×1500×1900mm|
|Humiture Range||(RT+10)~70°C; 30~98%RH|
|Humiture uniformity||± 2°C; ± 3%RH|
|Rainfall time||1~9999min, adjustable|
|Rainfall cycle||1~240min, interval is adjustable|
|Xenon lamp light||xenon long-arc lamp, 6KW±10%, 290-800nm, 1000W/m2±200|
|Sample spinning machine speed||1~5r/ min (adjustable)|
|Sample and xenon lamp distance||300~375mm|
|Box material||Stainless Steel SUS304#|
|Insulation materials||Superfine heat insulation cotton|
|Insulation body||Superfine glass fabric|
|Circulation system||Single cycle, centrifugal fan|
|Control system||Germany KLOCKNER-MOELLER progammable controler|
|Gas processing||dischager for fector and harmful gas|
|Power||AC380V, three phase five wire, 10KW|
|Use environment||5°C~+30°C, ≤85%R.H|
|Standard features||1. Filtering screen viewing window; 2.Moistureproof lamp|
|Security Protection||Security device for circuit break,|
|high pressure of the compressure, high temperature alarm|
US dry filters, solenoid valve Italy
Multi-wing strong air circulation blower, to avoid any dead ends, xenon lamp aging chamber allows a uniform distribution of temperature and humidity within the test area.
Air duct return air design cycle of wind, air pressure wind speed are in line with testing standards, and can open the door the moment the temperature and humidity stabilization time soon.
Heating, cooling, humidification system is completely independent can improve efficiency and reduce testing costs, increase life expectancy and lower failure rate
1. Xenon weathering test chamber wet ball gauze should be changed regularly: when the test surface is not clean cloth or harden, shall replace the measure
Test cloth. Test cloth for about three months to replace one, scrub with clean cloth when replacing body temperature, replacement of the test cloth at hand
It should be cleaned.
2. Check and adjust the xenon lamp weather resistance test chamber wet bulb water level
Water cylinder position is not too high, the water overflows, water cylinder or low absorbing cloth wet ball test is not normal, the impact of the wet bulb quasi
Accuracy, keep the water level is about six points can be filled.
Adjust the water level of the tube, adjust the level of the water box.
Contact Person: Mr. Liang