|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Packaging Details:||Strong Wooden Case|
|Delivery Time:||15 Days After Order|
|Payment Terms:||L/C, T/T, D/A, D/P, Western Union, MoneyGram|
|Supply Ability:||75 Piece / Pieces Per Month|
|Test Speed:||0.01-500mm/min (adjustable )||Screw:||High Precise Ball Screw|
|Test Area Material:||SUS304 Stainless Steel||Elongation Accuracy:||0.001mm|
|Power:||1∮, AC220V, 50HZ||Sensor:||Celtron Load Cell|
tensile testing equipments,
tensile strength tester
Force sensors Testing Machine 1000kg Capacity tensile strength testing machine
Force sensors Testing Machine 1000kg Capacity tensile strength testing machine is widely used in aviation, aerospace, petroleum chemical industry, machinery, metal materials and products, electric wire and cable, paper, printing and packaging, tape, bags handbags, textile fiber, pharmaceutical, food industry, etc. Testing for all kinds of materials and the physical properties of the products, semi-finished products, the choose and buy all kinds of different furniture can do tensile and compressive, the pull, bending, tear, peel, adhesion, shear test and etc.
The instrument configuration
Standard configuration: host, universal fixture, professional software, communication cable
The optional items: mini-printer, special printer cable, standard roller, test panel, sampling knife, floating roller fixture, non-standard fixture
Main components: frame, loading systems, measuring systems, data processing systems, fixtures and accessories, the most important are: load, measuring and data processing;
1. Universal Tensile Testing Machine structure is made of the paint-coated aluminium blanking plate. The interior is used the high-accuracy,low-resistance and zero clearance of two ball screw and oriented pole which improve loading efficience and structure rigidity.
2. Universal Tensile Testing Machine Use Panasonic seveo motor which ensure high efficiency,steady transmission and low noise. The accuracy of speed can be controled in 2%.
3. Computer Universal Tensile Testing Machine That Using the business computer as the main control mathine plus the special testing software of our campany can conduct all of testing parameter, work state, collecting data&analysis, result display and printing output.
Main Technical Parameter
|Stroke||800mm(not including fixture)|
|Test Speed||0.5~1000mm/min(control by keyboard inputing)|
|Test range||400mm MAX|
|Operation method||computer control|
|Motor||Panasonic servo motor|
|Accessories||customized clamps by appointed, force sensors,printer, and operation manual|
|Power||220V/50HZ or 110V/60HZ|
Stop way: the upper and lower limit safety set, the emergency stop button, strength and elongation elongation program set, the specimens are damaged sensor.
1) to find the body of the preliminary positive
2) tensile testing machine precision alignment
After pouring feet nut solids, before the cement is not dry, not allowed to tighten the nut and the feet start testing machine. (Cement drying time is generally less than 10-15 days) after the cement to be completely dry, tighten the nut luck feet on the installation accuracy of the testing machine for review, whether the alignment precision consistent. If they do not comply should be re-alignment.
Tensile testing machine during use, due to vibration test prone to loosening, so tensile testing machine used for a period of time, related parts should be tightened.
Speed tensile testing machine calibration:
1. First record the initial position of the cross arm of the machine, select the speed value on the control panel (using a straight ruler measuring standard cross arm stroke).
2. The starter while electronic stopwatch starts counting station for a minute, stopwatch arrival time while pressing the button to stop the machine, according to the stopwatch time record cross arm stroke value is the rate per minute (mm / min), observing the cross Tam stroke difference value and the ruler of the straight, and calculate the cross arm stroke error value should not exceed ± 1%.
Tensile testing machine Maintenance
1, machine parts, the appearance of regular cleaning to keep clean.
2, plating part please wipe oil to prevent rusting Kibo light.
3, part of the power screw and screw please add oil to protect sensitive transmission. (Add a little butter and oil mining lubricants can be mixed).
4, panel (ie, the control box display) wipe with a dry cloth to prevent wet, to avoid damaging IC electronic components.
5, the fixtures and fittings, please properly set, and remove anti-rust oil to prevent rust.
6, the work required to turn off the power.
Contact Person: Mr. Liang